Micro Dial Indicating Torque Screwdriver used for inspection and measuring of very small torque with precision.
Torsion bar mechanism makes accurate measurement possible.
With bi-directional scale plate, it can be used for retightening
Includes set of special Tohnichi bits
Applicable for international use including the EU region. Compliant with calibration procedures of ISO 6789 Type I Class D.
Wear safety glasses.
A tip of MTD bit is pointed. Wear a protective gear such as safety glasses.
ree from adhesion of lubricant such as oil and grease.
Failure to observe this may cause your hand to slip during tightening or measurement, resulting in an accident or injury.
No any cracks, scars nor rust.
Failure to observe this may cause damage, resulting in an accident or injury.
Do not run the torque screwdriver over the maximum allowable torque.
Exceeding the maximum torque may cause accuracy failure and damage.
Do not drop nor give a strong impact to the torque screwdriver.
This torque screwdriver is designed for precision measurement.
Use a suitable bit for a thread groove.
If the dimension and shape of the bit nose do not fit, the torque screwdriver may damage a tightening/measuring part such as a screw head.
Do not leave the torque screwdriver in a wet condition.
In case water is adhered, wipe it off immediately. Avoid storing the torque screwdriver in a high-temperature and humidity place. Failure to observe this may cause rust, resulting in accuracy deterioration and damage on the product.
Do not use the torque screwdriver in a dusty environment.
When dust enters inside, it may cause accuracy deterioration.
How To use
Replace MTD Bit
(1) Insert MTD Bit
Push up coupler collar, insert MTD bit deeply into the hole of the coupler center. Release the collar and it returns back itself by a built-in spring.
(2) Remove MTD Bit Push the coupler collar toward the scale plate to remove the bit. MTD bit comes off from the coupler center. When it's hard to remove, use a pliers to pull it.
Adjusting Zero Point
(1) Zero Point Setting MTD scale plate is designed rotatable. With holding coupler cover, turn the lower part of scale adjuster slowly to adjust MTD pointer correctly to 0, Zero.
(2) Presetting Called presetting method. Set the pointer to your target torque in advance. When tightening torque is reached to the target, the pointer is on zero. Holding the coupler cover, turn the lower part of scale adjuster slowly to adjust MTD pointer to the specified torque.
Holding MTD Torque Screwdriver
nip the knurled part of the case with a thumb and middle finger, and place an index finger on the top to hold the torque screwdriver. Operating thumb and middle fingers to measure torque.
Measuring Torque with a Special Jig
When making a special jig on your own, care should be taken to strike a balance between right and left torques. (In case of a horizontal shaft) It is convenient to manufacture the bit by cutting off the nose of the accessory bit.
We do undertake Servicing and repairing of Tohnichi MTD Indicating Type Torque Screwdriver TQC Sheen Range of products throughout india
Apart from supplying TQC Sheen Electrical Testing products in India through dealers network, catering to a variety of customers ranging from Atomic energy, Electricity Boards, Electrical Laboratory, Shopping Complexes, Shopping Mall, Electrical Contractors, Engineering Industries, Turnkey Construction Companies, Railways, Defence Installations, Research Laboratories, Educational Institutes and various Research and Development laboratories of leading national and multinational companies.
We DFT TECH, are TQC Sheen Authorized Distributors, Importers, Stockists, Dealers and Suppliers of Tohnichi MTD Indicating Type Torque Screwdriver Products in Chennai, Assam, Madurai, Coimbatore, Nagaland, Kerala, Manipur, Mizoram, Haryana, Sikkim, VIsakhapatnam, Tirchy, Hosur, Anjaw, Tumkur, Mysore, Kolkatta, Salem, Poonamalle and etc..